Fluke—37X Series Clamp Meters
What are the defects?
A screw connecting the printed circuit assembly to the test lead input jack can work itself loose and result in intermittent voltage readings, including a ‘0’ reading when voltage is present.
What are the hazards?
If the clamp meter is not reading correctly, users may be exposed to a potential electrocution hazard.
What should consumers do?
1. Stop using your affected 37X series clamp meter(s) even if you have not experienced problems.
2. Verify that your clamp meter is affected using the serial numbers provided in the table.
3. If the product falls within the serial number range, return the unit to Celemetrix for repair.
Electrical trade and retail outlets nationally
Recall advertisements and supporting documentation
Australian Competition and Consumer Commission is the responsible regulator for this recall.