Fluke—T2 Electrical Tester
What are the defects?
A Battery Contact To The Circuit Board Inside The Instrument Can Corrode Over Time Causing It To Operate Intermittently. When The Malfunction Occurs The Meter May Not Indicate That Voltage Is Present Placing The User In Potentially Hazardous Situation.
What are the hazards?
What should consumers do?
For Information Ring Philips Business Electronics On (02) 9805 4486. Units Will Be Serviced And Returned Directly To Customers At No Charge.
Australian Competition and Consumer Commission is the responsible regulator for this recall.