BiPAP A40 Ventilator
Serial numbers: V2887246677A0, V288724698F57, V28872472285C, V288724777FF1, V28872479968F, V288724808886, V28872485DF2B, V28872492B24C, V28872495C6F3, V288724981D16, V288725010D13, V2887251114CB and V2887251771FD
(Philips Electronics Australia Ltd - Positive airway pressure unit, bi-level)
What are the defects?
Certain devices were built with motor assemblies that may contain non-conforming plastic material.
What are the hazards?
If the non-conforming material is present patients may be exposed to:
1. Chemical exposure due to off-gassing, creating a potential biosafety or toxicological hazard.
2. Sudden failure of the device causing a Ventilator Inoperative condition, which may result in asphyxia if not immediately identified and addressed by the care provider.
What should consumers do?
Philips will replace all the affected devices, users will receive new A40 Pro units.
For more details, see https://apps.tga.gov.au/Prod/sara/arn-detail.aspx?k=RC-2022-RN-01118-1
If you believe your product is affected and you have not received a letter from Philips, please contact Philips Sleep and Respiratory Care by phone on 1800 009 579.
Traders who sold this product
Healthcare facilities in NSW, QLD, SA and WA