IBD Medical — Dario Blood Glucose Tracker App Android App Versions 4.3.0-4.3.2
What are the defects?
Potential issue with Android App versions of Dario Blood Glucose Tracker (4.3.0-4.3.2), regarding duplicate logging of a blood glucose level reading. The issue may cause blood glucose level readings that are transferred from the measurement screen to the 'Logbook' being replaced by an older blood glucose measurement and being displayed as a duplicate reading.
What are the hazards?
If inaccurate blood glucose level reading from the 'Data Entry' or 'Logbook' screens are used to determine the insulin dose or make treatment decisions, this may lead to incorrect dosing decisions, which can lead to serious health complications.
What should consumers do?
Update the latest Dario - Blood Glucose Tracker Android App version as follows:
Open the Google Play store
Search for 'Dario Blood Glucose Tracker' App
Press the 'Uninstall' button
Press 'OK' to confirm the question "Do you want to uninstall this app?"
Once completed, press 'Install' and verify 4.3.3 version (or above) has been installed by pressing the 'About this App' link.
For further information, contact IBD Medical on 02 8398 7520
Google Play Store
Therapeutic Goods Administration is the responsible regulator for this recall.