Serial Numbers Between 70521601 And 74165430. If The T2 Has A Serial Number Lower Than 70521601 And It Does Not Have An "r" Stamped After The Serial Number, The T2 Was Not Repaired In The November 1998 Recall And Must Be Returned To Fluke For Repair.
Reasons the product is recalled
The Positive Battery Contact Can Corrode Due To Exposure To Vibration And The Instrument Can Operate Intermittently. When The Malfunction Occurs, The Tester May Not Indicate That Voltage Is Present Placing The User In A Potentially Hazardous Situation.
The hazards to consumers
What consumers should do
Either Return With A Safety Notice Return Form Available By Faxing Fluke Australia (02) 8850 3300; Or Request A Free Field Repair Kit By Telephoning (02) 8850 3333 Or Faxing Fluke Australia.
Fluke Australia Pty Ltd