Item no. 2730641. Model No. ACD-10 PRO. Digital Clamp-On Multimeter. Item no. 2730652. Model No. ACD-10 TRMS-PRO. Digital Clamp-On Multimeter, True RMS. Item No. 2730665. Model No. ACD-14. HVAC Digital Clamp-On Multimeter. Item No. 2730676. Model No. ACD-14 TRMS. HVAC Digital Clamp-On Multimeter, TRMS.
Reasons the product is recalled
Both the ACD-10 and ACD-14 models "lock up" in certain circumstances failing to properly measure hazardous voltage levels on electrical systems. When the clamp meter locks up it provides readings under 1 volt or a millivolt reading. If the user does not follow industry standard practice by verifying the tester before and after checking the circuit, he/she may believe there is no voltage present when in fact there is.
The hazards to consumers
What consumers should do
Fluke Australia Pty Ltd will contact all customers.
Fluke Australia Pty Ltd
Where the product was sold